Get the complete project »
- The Complete Research Material is averagely 52 pages long and it is in Ms Word Format, it has 1-5 Chapters.
- Major Attributes are Abstract, All Chapters, Figures, Appendix, References.
- Study Level: BTech, BSc, BEng, BA, HND, ND or NCE.
- Full Access Fee: ₦4,000
Scientific evidence has revealed radiation as one of the causes of cancer and the use of radiation in medicine is on a continuous increase; studies have revealed that the medical use of x-ray is the major contributor of man-made source of radiation to the world‟s population. The drive for this study was the increase in the number of patients available for hysterosalpingography, HSG investigation. The aim of the study was to determine the radiation risks to the patients and staff during HSG procedure. A cross sectional study was conducted on 38 female patients in two diagnostic facilities in Katsina state of Nigeria.” It is important to know that the staff performing the procedure stays in the radiographic room together with the patient while the x-ray exposures take place, this should not be confused with the operator of the x-ray machine”. Quality control tests were conducted on the x-ray machines involved in the study as preliminary tests to ascertain the credibility of the generated data. The entrance surface doses ESDs to the patients‟ pelvis and anterior neck surface of the staff were measured by placing TLDs on the patients‟ pelvic surface and the shirt collar of the staff respectively. The mean measured patient ESD and staff ESD were 10.88mGy and 0.12mGy. The patients‟ mean measured effective doses to the ovaries, uterus, bladder, rectum, pelvic bone surface and the pelvic skin surface were 0.87mSv, 1.31mSv, 0.44mSv, 1.31mSv, 0.11mSv and 0.11mSv. While, the staff mean measured effective doses to thyroid, eyes and facial skin were 4.8µSv, 14.4µSv and 1.2µSv. The total exposed films used were between 4 to 6 films, this implies that the patients‟ ESD per film is 1.8 to 2.8mGy which is below the IAEA diagnostic reference level of 10mGy for Antero-posterior (AP) projection pelvis. And the mean radiation dose received by the staff is below the recommended occupational dose of a radiation worker by international commission on radiological protection ICRP 103. Though the mean measured effective dose values obtained were low, it does not indicate zero health effect since in radiation protection there is no safe dose for now. It should be understood in principle that dose acquisition has cumulative effect. Therefore, there is need for further optimization of procedure by reducing the number of patient radiation exposures (x-ray films taken), proper choice of exposure factors, tighter x-ray beam collimation and regular QC checks.
Keywords: Entrance surface dose, effective dose, hysterosalpingography HSG.
1.1 Background of the study
X-ray is a form of electromagnetic radiation which was discovered on November 8,
1895 by Wilhelm Conrad Roentgen. This ground breaking discovery gave birth to
radiography on December 27, 1895; when Roentgen took the first x-ray image of his
wife‟s hand (Explorable.com, 2010). This made scientists at that time use this amazing
ray to create images of the inside of the human body. But it was not for long when
people started complaining of skin burns and cancer. In the late 1896, Elihu Thomson,
an American physicist, deliberately exposed the little finger of his left hand to an x-ray
tube for several days, half an hour per day which resulted in pain, swelling, stiffness,
erythema and blistering.
In 1942, the leaders of the Manhattan District of U.S. asked Ernest O. Wollan, a cosmic
ray physicist at the University of Chicago, to form a group to study and control
radiation hazards and this was when the word „health physics‟ was born. However,
because of the various complaints that arose as a result of the uses of x-rays, various
bodies (both governmental and non-governmental) were created to provide solutions
and recommendations on how to reduce the adverse effects of radiation (Kathren &
Ziemer, 1980; Turner, 2007). .
Radiation has several applications; it is used in the industry, agriculture, medicine and
research. The medical use of radiation involves its use in conventional radiography,
fluoroscopy, angiography, nuclear medicine (scintigraphy), computed tomography,
mammography and radiotherapy.
- 1 -
The use of ionizing radiation in clinical practice is on the increase both locally (in
Nigeria) and globally. Certain checks have to be periodically done in order to ensure the
optimization and protection of patients from these dangerous rays; in addition to
investigate compliance level. „Survey data showed that physicians underestimate the
magnitude of radiation doses and their associated detrimental effects and thus
underestimating the level of risk to patients (Mercuri et al., 2011).
Measurements through quality control tests provide a means for setting and checking
standards to ensure good practice as an aid to the optimization of patients‟ protection.
Also estimates of the absorbed and effective doses to skin and tissues/organs in the
patient are needed to determine the associated risks. Data generated provide for
improvement in operations hence justifying techniques and procedures put in place in
Currently, the two fundamental principles (justification of practice and the optimization
of protection) have been placed under serious scrutiny in the achievement of ALARA in
the diagnostic radiology. In the light of the aforementioned, medical exposures are
receiving increasing attention with the realisation that clinical x-rays make the major
contribution from manmade sources to the population collective dose; also, there is
considerable potential for reducing doses without detriment to patient care (IRPA9,
Hysterosalpingography (HSG) is a special radiographic examination of the uterine
cavity and the fallopian tubes using radio-opaque dye for the evaluation of infertility in
women. „It involves unavoidable irradiation to the ovaries of women in child bearing
age; therefore, optimization is required in all involving procedures to ensure reduction
in the probability of the associated risk‟ (Suleiman et al., 2008). With the growing
concern about radiation doses received by patients and cancer incidence over the years,
- 2 -
there has been an emergent requirement for information on typical doses and the range
of doses r
You either get what you want or your money back. T&C Apply
You can find more project topics easily, just search
SIMILAR PHYSICS FINAL YEAR PROJECT RESEARCH TOPICS
» ABSTRACT In this study of the physics of stars and their astronomical identification, it is evident that stars undergo certain physical processes as t...Continue Reading »
2. EFFECTS OF PRACTICAL THEORY ARRANGEMENTS ON THE ACHIEVEMENT OF PHYSICS STUDENTS’ IN THE CONCEPT OF SIMPLE PENDULUM IN SECONDARY SCHOOLS IN EKET LOC...» Abstract The study aimed at comparing effects of practical theory arrangements on the achievement of physics students in secondary schools in Eket Loc...Continue Reading »
» ...Continue Reading »
» ABSTRACT This project is a construction of a module of a zener diode and how output voltage varies with increase in current in forwards bias using mil...Continue Reading »
5. ASSESSMENT OF OCCUPATIONAL EXPOSURE AND RADIATION PROTECTION IN SOME SELECTED WELL-LOGGING AND INDUSTRIAL RADIOGRAPHY FACILITIES IN NIGERIA» ABSTRACT Nigeria has for a very long time engaged in the peaceful application of nuclear technology. The use of ionizing radiation due to its unique p...Continue Reading »
6. A STUDY ON THE COMPETENCY OF PHYSICS STUDENTS IN THE USE OF INFORMATION COMMUNICATION TECHNOLOGY A CASE STUDY OF UNIVERSITY OF BENIN» ABSTRACT This study investigates the competency of physics students in the use of information and communication technology in University of Benin (UNI...Continue Reading »
» CHAPTER ONE INTRODUCTION Background to the study Astrophysical jets are observed in the Universe in a large variety of environments and under a wide r...Continue Reading »
» ABSRACT This paper work contains details of the design and construction of a remote control fan regulator. The system simply consists of an infrared (...Continue Reading »
» CHAPTER ONE INTRODUCTION 1.1 INTRODUCTION Fuel burn up was shown to be linearly dependent on the reactor thermal power (Podvratnik, 2011). It is there...Continue Reading »
» ABSTRACT Wi Fi signals are susceptible to signal loss as they travel hence Wi Fi users are rarely completely satisfied with the signal strength offere...Continue Reading »